Unstretched: 65-90 cm
When sufficiently stretched for use: 74-115 cm
The Beat2Phone ECG sensor uses a rechargeable battery that is charged in the included Beat2Phone ECG charging connector.
The battery charges from empty to full in approximately one hour.
In some mobile devices, the device location must be turned on to connect to the sensor.
From the Beat2Phone ECG application settings, check that your sensor is connected to the mobile device. Try disconnecting the sensor by choosing “Forget sensor” and connect to the sensor again. If this does not resolve the issue, try rebooting the mobile device Bluetooth connection and the Beat2Phone ECG application. If still not resolved, try rebooting your mobile device. Finally, reboot the sensor by briefly charging it.
From 24 hours up to 48 hours if the sensor battery is fully charged.
Many mobile devices have, by default, a rather aggressive battery consumption optimization, which can turn off applications that appear inactive. This battery optimization sometimes turns off the Beat2Phone application, which causes the measurement to stop. Usually, this optimization can be adjusted from the mobile device settings, depending on the model of the device. The most reliable measurement can be performed when the mobile device is not used for anything else during the measurement. Even more reliable performance can be achieved if the application remains active, and the mobile device is being charged. To make sure that the measurement has not stopped, it is recommended to check the status of the measurement from time to time.
The Beat2Phone ECG application has not been tested on all mobile devices.
It has been found to work well in the following devices:
In the following devices, Beat2Phone does not work properly (Bluetooth problem):
You can find the short user guide from the instructions for use page >>
You can find instructions for use from the instructions for use page >>